View and Download Fluke datasheet online. Fluke and Geo Earth Ground Testers Datasheet. Accessories pdf manual download. Also for. View and Download Fluke user manual online. Earth/Ground Tester. Measuring Instruments pdf manual download. The new Fluke and GEO Earth Fluke Corporation 16 3 and 16 5 GEO Earth Ground Testers manual—all inside a rugged Fluke carrying case.
|Published (Last):||10 August 2006|
|PDF File Size:||9.50 Mb|
|ePub File Size:||14.73 Mb|
|Price:||Free* [*Free Regsitration Required]|
Test Equipment Depot – Specifications are subject to change without notice. All product names are trademarks of their respective companies. Fluke authorized resellers shall extend this warranty on new and unused products to end-user customers only but have no authority to extend a greater or different warranty on behalf of Fluke.
Fluke Basic GEO Earth Ground Tester Kit
Warranty support is available only if product is purchased through a Fluke authorized sales outlet or Buyer has paid the applicable international price. Table of Contents Title Page Introduction Introduction The Earth Ground Tester Tester or Product is a compact, field-rugged manusl that performs all four types of earth ground measurement.
Specifically, the Tester is able to measure earth ground loop resistances using only clamps — called Stakeless testing. A Caution identifies conditions and procedures that can amnual damage to the Product or the equipment under test.
Do not dispose of this product as unsorted municipal waste. Storage If the Tester is stored for an extended period of time or is not in use for a long time, you The transformer has a transformation ratio between 80 and This enables the user to measure on high voltage pylons without separating the overhead earth wires or earth strips at the bottom of the pylons. See Table 4 for location and description of each display element.
If you have problems, see How to Troubleshoot.
Measurement values are shown on a liquid crystal display with correct decimal point and unit. Additional special characters indicate measurement mode, operating condition, and error messages.
The Tester also makes measurements with an external current transformer, with which a measurement of single resistance branches in 16223 networks lightning protection and high voltage pylons with cabling can be performed without separating parts of the system.
Select 4 POLE function. Earth ground stakes are not used.
A known voltage is induced by one flke, and the current is measured with the second clamp. Stakeless Ground Loop Measurement One part flows via pylon construction to the overhead earth wire and further to the parallel circuited pylon earthing resistances.
Read out measured value R Note Before setting the earth stakes for probe and auxiliary earth electrode make sure that the probe is set outside the potential gradient of earth electrode and auxiliary earth electrode.
Fluke 1623-2 User Manual
The measuring procedure shown in Figure 10 uses the method developed by Wenner F. Wenner, A method of measuring earth resistivity; Bull. Measurement of Soil Resistivity 1. See Figure 11 for steps Troubleshooting Step Description External voltage Uext too high If the external voltage applied to the instrument is too high, usually from flkke currents in the system under test, no measurement manula be started see Specifications for Uext limit.
If the meter still fails manuual operate properly, pack it securely in its original container if available and forward it, postage paid, to the nearest Fluke Service Center. Include a brief description of the problem. Fluke assumes NO responsibility for Current and voltage measurement Measurement voltage: Also See for Quick reference manual – 2 pages. Page of 40 Go. Table of Contents Add to my manuals Add.
Fluke 1623 Basic GEO Earth Ground Tester Kit
Page 2 Fluke authorized resellers shall extend this warranty on new and unused products to end-user customers only but have no authority to extend a greater or different warranty on behalf of Fluke. Page 4 Users Manual Page 6 Users Manual Page 8 Users Manual Introduction Introduction The Earth Ground Tester Tester or Product is a compact, field-rugged instrument that performs all four types of earth ground measurement.
Safety Information Users Manual Safety Information A Warning identifies hazardous conditions and procedures that are dangerous to the user. Advanced Operation – allows measurements of earth electrode resistances without disconnection of the earthing system or disengaging the overhead earth wire.
Page 30 Users Manual As all four pylon stubs are connected to the foundation earth of the pylon, the measuring current I is divided into five components according to the present resistances meas involved.
Maintenance If the meter still fails to operate properly, pack it securely in its original container if amnual and forward it, postage paid, to the manhal Fluke Service Center. Page 40 Users Manual Comments to this Manuals Your Name.
Enter text from picture: By pressing ‘print’ button you will print only current page. To print the manual completely, please, download it.
FLUKE USER MANUAL Pdf Download.
Don’t show me this message again. Got it, continue to print. Don’t have an account?